High negative pressure, large flux, low noise.
Safe to Use: The machine does not produce a positive pressure to ensure that its performance is reliable, safe to use.
1000ML Water Tank: This item reliefs in less than a minute, 1000ml high capacity water tank for less refills.
Portable: Small size, light weight, easy to carry, especially to adapt to a variety of emergency situations and travel needs.
Power:AC 110V 60Hz
The input power:90VA
The limit negative pressure:≥ 0.06MPa
Negative pressure adjustment range:0.02MPa to the limit negative pressure
Pumping rate:≥ 15L / min
Fuse tube:F1.5AL250V,φ5 × 20
Noise:≤ 65dB (A)
Dimensions:370 × 180 × 285 (mm)
2*Suction catheter (Adults and children)
1*Storage bottle (1000mL)
Inhale the water before sucking to moisten the inner wall of the suction tube and the latex tube to prevent the sputum from adhering to the inner wall of the tube and causing cleaning trouble.
The end of the suction tube was wetted with physiological saline before sucking and then sucked.
Do not take more than 15 seconds each time, if you use it again, it should be separated for 3 minutes to prevent damage to the visceral mucosa.
If the patient’s visceral mucosa is inadvertently sucked during sucking, the suction tube should not be taken out at will. The suction tube should be opened in time to release the negative pressure. After the patient’s discomfort disappears, the suction tube can be taken out.